In this demonstration we measure the standing wave ratio (SWR) of a load, in this case an UHF antenna. A constraint is that we want to do this in a cost-effective way up to 1.7 GHz. My earlier discussed VNA is not capable of doing this, you can get VNAs that can but they rapidly become rather expensive.
In the demonstration video we generate noise with a tool that essentially amplifies the noise of a zener diode, you could make it yourself, but if you google for BG7TBL noise generator then you will soon find out it is easier to order it than to build it yourself. Noise from the BG7TBL tool has the property that it is random phase by frequency and that the amplitude is somewhat constant over the entire spectrum. Put it on the scope and you get what you expect: something that resembles white noise. The tool needs up to 12 Volt to operate, it will get hot, but it also works on a 9 Volt battery.
If you insert noise in the SDR (I used my airspy) then it will fill the entire HF VHF and UHF spectrum, and you may wonder why you would like to do that at all. Please insert attenuators because the output of the BG7TBL tool is rather high, so I added 30 dB attenuation before I did the measurement with the SDR. When I measure the reflection properties of a load then 10dB goes before the bridge, and 20 dB goed after the bridge to prevent that the bridge is influenced too much.
There is a very good reason why you would like to insert artificial noise into your SDR. The reason is that it allows you to measure the performance of filters or loads connected in between the source and the receiver. In the video we inserted a RF bridge between the source and the receiver.
The RF bridge comes from the transverter store on ebay, see the movie, also there the conclusion is, it is easier to buy one than to make one. The bridge has an input, an output, there is a reference port and there is a device under test port. There are only SMA connectors on the PCB, the schematic is as follows:
On the input (left in the schematic) you insert a signal into the RF bridge, and on the reference port (top port in the schematic) you insert a nominal dummy load of 50 Ohm. On the output of the RF bridge (right port in the schematic) you measure the throughput with your SDR (an airspy). There are now several possibilities depending on what you connected to the DUT port (bottom port in the schematic):
- Nothing is connected to the DUT port, in this case the signal reflects back into the RF bridge and this is picked up by the receiver. The throughput is called the directivity of the RF bridge. (for details see the discussion on the transverter-store web page).
- The DUT port has a dummy load of 50 Ohm (or the same amount of Ohm as the reference port). In this case you will notice that no signal will pass the bridge and that everything is dissipated in the dummy load at the DUT port.
- Connect an unknown load to the DUT port, and now you will see that some of the signal is reflected back to the receiver, but that the rest of the signal is dissipated into the load on the DUT port.
With the first and the second measurement you calibrate the set-up, with the third measurement you can see who much dB is dissipated in the unknown load. You can measure this in the waterfall plot, and hence you can measure the so-called reflection loss RL in decibel caused by the unknown load. Next you need to convert the RL into a SWR value, the equation is as folows:
RL = 20 log10( (SWR-1)/(SWR+1) )
I measured on the waterfall a reflection loss of 20 dB near 440 MHz, this is compatible with a SWR of 1.2, the connected antenna is a miniature version of a 70 cm amateur band antenna.
Last update: 23-5-2018